L utilisation d’une électrode à anneau tournant a permis d’obtenir la totalité de la relation J(E) entre la densité de courant J et le potentiel électrode tournante. ETUDE VOLTAMMETRIQUE DE LA REDUCTION DES IODATES A UNE ELECTRODE TOURNANTE DE PLATINE Par J. BADOZ-LAMBLING et C. GUILLAUME. Chem Instrum – Margarit J, Lévy M () Étude theéoretique d’une électrode tournante á double anneau. Partie I. Recherche du facteur d’efficacité.
|Published (Last):||7 August 2010|
|PDF File Size:||12.77 Mb|
|ePub File Size:||14.9 Mb|
|Price:||Free* [*Free Regsitration Required]|
Apply a small amount of a silicon-based grease to the top of the shaft before installing the shaft into the motor coupling.
Section 4 — Operation. Always turn the rotation rate control knob completely counterclockwise towards the zero rotation rate position before turning on the rotator.
Use extreme caution when operating the rotator at rotation rates above.
The lower pair of brushes blue contacts the ring electrode while the upper pair red contacts the disk electrode. It is important to understand that the low humidity environment tournwnte in most glove boxes increases the rate of wear on both the brush contacts and the internal brushes within the toudnante itself. Both of these lines must be connected to the rotating electrode brushes. Each shaft is specially designed to provide one or two current paths down to the electrode tip.
The chassis of the motor unit is also normally connected to the chassis of the control unit, and thus, to earth ground. An insulating spacer just below the mounting area isolates the mounting area from the electrode contact area. The drive line carries current while the sense line measures the potential. Inspect the shaft to be certain that it is not damaged. Smaller cells may be clamped using a traditional laboratory clamp secured to the center port see Figure 4.
There is no universally accepted color coding scheme for marking potentiostat cell cable connections. There are two important grounding connections that should be considered in any rotating electrode experiment. The lower pair of brush contacts are only used with a rotating ring-disk electrode see Figure 4. When raising and lowering the motor unit along the main support rod, be sure to hold the motor unit carefully so that it does not unexpectedly fall and break the glass cell located below the motor unit.
The lower portion of the shaft below the lower electrofe spacer provides electrodw contact with the ring electrode. Some potentiostats offer both options, and if this is the case, only one of the two options should be used at a time, preferably the ellectrode terminal.
Such a shaft is either improperly installed or physically damaged. Do not use or attempt to rotate an electrode tip that has been dropped or otherwise physically damaged. It is possible that the brush assemblies on a rotator that has been in elecyrode for some time may have been replaced or swapped, and thus, the colors of the brushes may not be as described in the previous paragraph or as shown in Figure 4.
In addition, the vertical position of the motor unit is easily adjusted. If desired, a jumper setting inside the control box can be configured to use the opposite logic see Section 6. Electrical connection is accomplished using one or more silver-carbon brushes to contact metal surfaces on the upper portion of the rotating shaft.
Usually, it is easier to mount and clamp the cell in a fixed vertical position. Turn off the rotator, disconnect electrical power, and remove the electrode tip immediately. Newer rotators have a shielded motor control cable which has HD connectors on each end of the cable. To avoid issues with signal tuornante when making electrochemical measurements, it is important to properly ground all metal objects near an electrochemical cell to the earth ground. Use a short banana jumper cable to connect the opposing brushes together see Figure 4.
Appareils d’Analyse et Mesure en Electrochimie – OrigaLys
Text of the Claims and Abstract are posted: The first important connection is to ground any metal clamps or cell supports near the electrochemical cell see Figure 4. The upper red pair of brushes makes contact with the disk or cylinder while the lower blue tpurnante makes contact with the ring on a rotating ring-disk electrode.
Thank you for waiting.
After the cell has been mounted and the electrode has been lowered into the cell, securely mount the enclosure by hooking the enclosure to the four pins on the enclosure base see Figure 4. Even with the knob in this position, there may be some residual rotation typically less than in either the clockwise or counter-clockwise direction. Users wishing to rely electeode this information should consult directly with the source of the information.
Position the motor unit with respect to elsctrode glass cell so that the electrode tip is immersed approximately into the test solution.
The power switch on the back panel also acts as a circuit breaker to help protect the control unit circuitry. For a cell with multiple side ports, carefully orient the cell so that any accessories mounted in the side ports have enough clearance. When properly mounted in the rotator, the upper of the shaft is inside the motor unit, while the remaining length of the shaft extends down below the motor unit. Do not use an electrode electorde which appears to wobble, vibrate, or tilt away from the axis of rotation while rotating.
The cable which connects the reference electrode to the potentiostat should be of the electrkde coaxial type, and care should be taken to route this cable well away from noise sources such as power cords, networking cables, or video monitors. The enclosure has slots along the bottom of the window that provide clearance for routing cell cables and any tubing out of the enclosure.
These current paths are electrically electgode from the mounting area near the top of the shaft. Two clamshell doors surround the tournate chamber.
Do not turn on the rotator or rotate the electrode shaft if the shaft is not securely mounted in the motor coupling. If the cell cables from the potentiostat also terminate with banana plugs, then these plugs can simply be inserted directly into either end of the jumper cable. The uppermost portion of the shaft is used to mount the shaft into the rotator see Figure 4.
The analog rate tournabte signal from the potentiostat is a voltage electrodw is proportional to the desired rotation rate.